HART Test System
Current Released Version Release 3.5 - October 2017The following CRs were resolved in this release: CR NumberSummaryAffected Components3630 Does...
Wed, 24 Oct, 2018 at 9:37 AM
Please download the attached Data Sheet (PDF).
Wed, 22 Mar, 2017 at 10:26 AM
Running tests on the HART Test System 1. First connect a registered RS-232 HART modem from the serial port labeled RS-232 (for black test ...
Fri, 22 Jul, 2016 at 12:35 PM
Please note that many companies manufacturer RS-232 modems but only the HART-Registered modems are guaranteed to have passed the HART Protocol standards a...
Tue, 1 Aug, 2017 at 12:28 PM
HSniffer replaced ANALYS in HART Test System version 3.0. HSniffer runs on the same unit as the test system.
Tue, 24 Nov, 2015 at 4:55 PM
Are tests beyond CAL074D in the Common Practice Command test specification supposed to be manually tested?
FieldComm Group does not have standard test automation for CAL079, CAL080, CAL091, CAL115, and CAL512. Consequently the test administrator uses a Generic De...
Tue, 1 Dec, 2015 at 4:35 PM
Please file a support ticket and attach all relevant log files and screen shots to describe your issue.
Tue, 1 Dec, 2015 at 5:03 PM
Currently test automation does not exist for CAL001, CAL060, CAL063, CAL070, CAL064, CAL065, CAL066, CAL068, CAL069, CAL062,CAL079, CAL080, CAL107, CAL038...
Thu, 9 Nov, 2017 at 12:51 PM
If the slave does not respond to a message it is considered an error. If the slave does not respond to a retry it is another error. If the slave answers the...
Fri, 22 Jan, 2016 at 10:39 AM
I have an older version of KIT-192, how do I update my HART Test System to the latest released version?
Open a terminal on the HART Test System, type “kit192ver” at the prompt. Note the version number reported on the screen. If the current version...
Thu, 8 Feb, 2018 at 10:30 AM